Please use this identifier to cite or link to this item:
|Title:||Microfluidic rheometer based on hydrodynamic focusing||Authors:||Nguyen, Nam-Trung
Yap, Yit Fatt
|Keywords:||DRNTU::Engineering::Mechanical engineering||Issue Date:||2008||Source:||Nguyen, N. T., Yap, Y. F., & Sumargo, A. (2008). Microfluidic Rheometer Based on Hydrodynamic Focusing. Measurement Science and Technology, 19(8).||Series/Report no.:||Measurement science and technology||Abstract:||This paper reports the concept and the optimization of a microfluidic rheometer based on hydrodynamic focusing. In our microfluidic rheometer, a sample stream is sandwiched between two sheath streams. The width of the middle stream depends on the viscosity ratio and the flow rate ratio of the liquids involved. Fixing the flow rate ratio and using a known Newtonian liquid for the sheath streams, the viscosity and the shear stress of the sample stream can be determined by measuring its width and using a prediction algorithm that uses the known channel geometries, fluid properties, flow rates and the focused width as input parameters. The optimization reveals that a measurement channel with a high aspect ratio is more suitable for a sample liquid with viscosity higher than the reference value. For a sample liquid with viscosity of the same order of magnitude or lower than the reference, a low aspect ratio is more suitable. Furthermore, the measurement range and the relative error can be improved by adjusting the flow rate ratio between the core stream and the sheath stream. A microfluidic device was fabricated in polymethylmethacrylate (PMMA). Using this device, viscosities of deionized (DI) water and polyethylene oxide (PEO) solutions were measured and compared with results obtained from a commercial rheometer.||URI:||https://hdl.handle.net/10356/98121
|DOI:||10.1088/0957-0233/19/8/085405||Rights:||© 2008 IOP Publishing Ltd. This is the author created version of a work that has been peer reviewed and accepted for publication by Measurement Science and Technology , IOP Publishing Ltd. It incorporates referee’s comments but changes resulting from the publishing process, such as copyediting, structural formatting, may not be reflected in this document. The published version is available at: [http://dx.doi.org.ezlibproxy1.ntu.edu.sg/10.1088/0957-0233/19/8/085405].||Fulltext Permission:||open||Fulltext Availability:||With Fulltext|
|Appears in Collections:||MAE Journal Articles|
Items in DR-NTU are protected by copyright, with all rights reserved, unless otherwise indicated.